Universal whole body phantom
The Whole Body Phantom uses for calibration of a whole body counters (WBC) for measuring the activity of radionuclides incorporated in human body with use of scintillation and semiconductor detectors of gamma-radiation in the energies range from 50 to 3000 keV.
The Phantom can be used in radiation medicine centers, nuclear power stations, research institutes, industrial hospitals and other institutions performing radiation monitoring of personnel and population, as well as by the Federal Agency for Technical regulation and metrology in order to comply with the Traceability Law.
Universal whole body phantom, adequate on interaction with gamma-radiation to biological soft tissue of the human body, containing uniformly distributed radionuclide.
Universal phantom is a collapsible construction and includes a set of components:
- Polyethylene blocks of a type “solid” (90pcs.) and “half” (40pcs.);
- Rod radionuclide sources of a type “nominal” (180pcs.) and “half” (80pcs.);
- Connectors (280pcs.);
- Fixings (10pcs.);
- Mounting blade;
- Chest (with left and right lungs) and neck (with thyroid glands) phantoms.
There are six types of the whole body phantom assembly with age and anthropometric characteristics given in table, for the three standard measurements geometries on whole body spectrometer: standing (lying), sitting and sitting bended.
The whole phantom with radionuclide sources forms a reference sample of activity of incorporated radionuclides uniformly distributed in the human body. This reference sample reproduces the gamma-radiation spatial-energy spectrum of incorporated radionuclides to transfer the activity value of radionuclides to whole body spectrometers by means of its calibration.
Rod sources are made in the form of the sealed radioactive sources, which eliminates the intake of radionuclides into the environment by transportation, storage and use. Assembly of polyethylene blocks without the radionuclide sources forms the background samples (background phantoms) of the appropriate size.
The chest phantom is the construction from the polyethylene (polyurethane for Am-241) units, which are the simulators of the adult man thorax. There is the polyurethane unit inside of the thorax simulator which is the lung simulator (2pcs). The radionuclides are distributed uniformly in the lung simulator.
Standard values of activity
The whole body phantoms can be provided for rent.
This phantom can be calculated using the MCC-MT software.